Any equipment has been used for a long time, and there will be some problems more or less, so the electronic tensile testing machine is no exception. After a long time, there will always be some minor problems. Let Cable Detection Equipment Manufacturer tell you some solutions to the problems that I often encounter.
1. If the test force and displacement do not change during the test, check whether the connection data line between the computer and the Electronic Tensile Test Machine is loose or loose (if these loose or fall off, install and connect and tighten);
2. Check if the connected sensor is correct (whether the sensor is too large or too small, whether it is damaged); check whether the sensor has been hit when the test or keyboard is operated recently;
3. The power of the test machine host is not bright and cannot be moved up and down. Check whether the power supply line connected to the test machine is connected properly; check whether the emergency stop switch is in the screwed state; check whether the power supply voltage of the test machine is normal; check whether the fuse on the machine socket is blown, please remove the spare fuse and install it. .
4. Check whether the calibration or calibration function of the software is used before the problem occurs; check whether the calibration value, calibration value or other information in the hardware parameters are manually changed (if there is no calibration reset, it must be calibrated and reset)
Copyright © Xingtai Longjia Electronics Technology Co.,Ltd. All Rights Reserved | Sitemap |Powered by